Presure, Volume and Temperature are measured at once. The volume is read from the syringe and the temperature and pressure are displayed on Data Studio. As you vary volume you can see the how pressure and temperature react.
There is a white stop built into the syringe. It stops at 20 mL.
A small amount of air in the tubing causes an error. Use the formula (V1+V0)/(V2+V0) =P2/P1 where V0 is the air in the tube.
There is a response time of about half a second between the pressure and the temperature measurement.
CI-6527A Thermistor Sensor CI-6532A Pressure Sensor-Absolute